N5183A LABVIEW DRIVER
I put the plugin in another mainframe with the same output: I am triying to set continues frequency range from eg: Is this a bug in the driver? I am switching Sync on and off to trigger DVM If you do not need this level of detail, you can find the complete programming examples in.
|Date Added:||21 February 2013|
|File Size:||26.77 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Please help me by sharing example code.
Signal Generators and Sources | Keysight Community
The VI has a ring constant that allows you to change what labveiw first check is looking for. No matter which programming environment you use to create the waveform data, make sure that the data conforms to the data requirements shown on.
You can use various programming environments to create ARB waveform data. This is done to help you understand the code algorithm in creating the I and Q data, so you can leverage the concept into your programming environment. I put the plugin in another mainframe with the same output: Net, Labview, and so forth. I labviww the above error when trying to enable the output. I followed all the instructions provided by the “help” section. It took a little digging.
Creating waveform data – Agilent Technologies N5183A MXG User Manual
The primary 5n183a was that the NB does not support modulation and the configure VI was attempting to send a modulation related command. How did you get the labvieww VI to check for the presence of the modulator? Hardware setup is the one following: The waveform needs to be syncronized with an external trigger. Is there a SCPI command for this? I am triying to set continues frequency range from eg: Get a feed of this content Use this view in a tile. I switched it to the MXG-B series and that fixed the problem.
I am switching Sync on and off to trigger DVM Most Active Software Boards: At the command prompt, type: The ultimate fix was that the “Configure RF Output” VI runs labvie series of checks up front to determine the type of instrument that is connected.
I’m using a NB to generate MHz pulses, about 1us wide and 10us apart, and viewing a pulse on an oscilloscope. If I’m going to use to signal generator for differenti I’ve noticed that for an arbitrary pulse period say 9 or 11usthe phase jumps around for one pul Accepted by topic author john Hello, I want to apply a differential signal input to my rectifier circuit.
If the first check returns a true, then the modulation command is not added to the string.
Is this normal and expected? I want to use NB vector signal generator with UP power distributor network.